Avery Weigh-Tronix 640 Indicator Calibration
Watch this video to calibrate your indicator.
- Press and hold the "Hold/Menu" button for three beeps then release.
- Enter "640". Use "RM" to increase the displayed value and "M+" to decrease it. Press the "Hold/Menu" button to move one digit to the left.
- Press "Print/Select" to accept and "640" will be shown again.
- Press "Print/Select" and "CONFIG" will be displayed.
- Press "Print/Select" once more and the config number will be displayed.
- Enter a new config number using "RM" to increase each digit and "HOLD/MENU" to move one digit to the right. 98300 is a custom config number that should work for most applications.
- Press "PRINT/SELECT" to enter the number and return to "CONFIG".
- Press "HOLD/MENU" and "CUSTOM" will be displayed.
- Press "Print/Select" once more and the custom number will be displayed.
- Enter a new CUSTOM number using "RM" to increase each digit and "HOLD/MENU" to move one digit to the right. This number is equivalent to a digistar calibration number and is unique to the specific application.
- Press "PRINT/SELECT" to enter the number and return to "CUSTOM".
- Press "NET/GROSS" repeatedly to return to the gross weight screen.
Avery Weigh-Tronix Load Cell Calibration Numbers
The reading @ 0.4mV/V can be used as the Calibration number for a Digi-star indicator, or the Custom number on a Weigh-tronix indicator.
Avery Weigh-Tronix 1040 Indicator calibration
Push and hold the MENU key until it says “SETPAS”. (*Note: If the indicator just sits there and beeps after 3-4 seconds, let go of the MENU key, at that point “SETPAS” should appear) Type 1040 Once “1040” is on the screen press SELECT. (*At this ...
Changing Setup and Calibration Numbers in a Digi-Star Indicator
Changing Setup and Calibration Numbers in a Digi-Star Indicator 1. Hold the ZERO button and the ON button until "SETUP" is shown on the screen. 2. After "SETUP" the indicator will display its current setup number with the last digit flashing. A good ...
Calibration Number Equation
Here is the equation to figure out what the calibration number would be. .4 mv/v X Capacity X # of Bars / mv/v Output of the weigh bar = good starting Calibration # Example: We will be using a 141907 weigh bar in this example. ...
GT400 and UNVERFERTH 410 Calibration
GT400 and Unverferth 410 Calibration Video