Avery Weigh-Tronix Load Cell Calibration Numbers
The reading @ 0.4mV/V can be used as the Calibration number for a Digi-star indicator, or the Custom number on a Weigh-tronix indicator.
640 Indicator Calibration
Watch this video to calibrate your indicator. Press and hold the "Hold/Menu" button for three beeps then release. Enter "640". Use "RM" to increase the displayed value and "M+" to decrease it. Press the "Hold/Menu" button to move one digit to the ...
Avery Weigh-Tronix 1040 Indicator calibration
Push and hold the MENU key until it says “SETPAS”. (*Note: If the indicator just sits there and beeps after 3-4 seconds, let go of the MENU key, at that point “SETPAS” should appear) Type 1040 Once “1040” is on the screen press SELECT. (*At this ...
Changing Setup and Calibration Numbers in a Digi-Star Indicator
Changing Setup and Calibration Numbers in a Digi-Star Indicator 1. Hold the ZERO button and the ON button until "SETUP" is shown on the screen. 2. After "SETUP" the indicator will display its current setup number with the last digit flashing. 3. ...
GT400 and UNVERFERTH 410 Calibration
GT400 and Unverferth 410 Calibration Video
Entering a Configuration Number on a Model 715 Indicator
Entering a Configuration Number on a Model 715 Indicator This is the best method to calibrate a 715 if using AWT loads cells of a known calibration diameter. 1. Type in 715 2. Hold the hidden E key for two seconds until 715E is displayed. ...